# Sensors for Mechatronics || Inductive and Magnetic Sensors

• View
213

• Category

## Documents

Embed Size (px)

### Text of Sensors for Mechatronics || Inductive and Magnetic Sensors

• 6 Inductive and Magnetic Sensors

Inductive sensors employ variables and parameters like magnetic induction B, mag-

netic flux , self-inductance L, mutual inductance M or magnetic resistance Rm. Bya particular construction of the device, these quantities are made dependent on an

applied displacement or force. First we review various magnetic quantities and

their relations. Next the operation and the specifications of the major types of mag-

netic and inductive sensors are reviewed. Special attention is given to transformer-

type sensors. The chapter concludes with a section on applications.

6.1 Magnetic and Electromagnetic Quantities

6.1.1 Magnetic Field Strength, Magnetic Induction and Flux

The magnetic field strength H, generated by a flow of charged particles, is defined

according to

I5

IC

HUdl 6:1

where I is the current passing through a closed contour C (Figure 6.1A). The quan-

tities H (A/m) and dl (m) are vectors. For each configuration of conductors carrying

an electric current the field strength in any point of the surrounding space can be

calculated by solving the integral equation (6.1). A current I through a long,

straight wire produces a magnetic field with strength H5 I/2r at a distance r fromthe wire. So the field strength is inversely proportional to the distance from the

wire. The field lines form concentric circles around the wire so the vector direction

is tangent to these circles (Figure 6.1B).

Only for structures with a strong symmetry, simple analytical solutions can be

obtained. Magnetic fields of practical devices and shapes are studied using FEM

(finite element method). The space is subdivided in small (triangular) areas and for

each area the equations are solved numerically. FEM programmes calculate the

field strength and direction over the entire region of interest; the results are visual-

ized in colour or grey-tone pictures or with field lines.

As an illustration Figure 6.2 shows the magnetic field of a permanent magnet

and of a wire loop with a DC current. The FEM programme also shows the numeri-

cal values of the field quantity in each point of the space enclosed by a specified

boundary.

Sensors for Mechatronics. DOI: 10.1016/B978-0-12-391497-2.00006-6

http://dx.doi.org/10.1016/B978-0-12-391497-2.00006-6

• Evidently a stronger field strength can be obtained by increasing the current.

However a more efficient method is to make multiple turns of the wire. Each turn

carries the current I, thus contributing to the field strength. For example the field

inside a coil with n turns is proportional to the product of n and I. The product n I(expressed as ampere-turns) is a measure for the strength of such a magnetic

source.

Other quantities that describe magnetic and induction phenomena are the mag-

netic induction B (unit Tesla, T, kg/As2) and the magnetic flux (unit Weber,Wb5 kg m2/As25Tm2). By definition the flux is:

5ZZ

S

BUdA 6:2

In words the flux is the inner vector product of the magnetic induction vector

and a surface patch dA, integrated over the total area S for which the flux is

Figure 6.2 Examples of magnetic fields obtained by FEM analysis: (A) a permanent magnet

and (B) a single turn with DC current 1 A.

Figure 6.1 (A) Magnetic field generated by current I, (B) calculation of field strength due to

a straight wire carrying a current I and (C) calculation of magnetic flux.

126 Sensors for Mechatronics

• calculated (Figure 6.1C). In particular for a homogeneous field that makes an angle

with the normal on a flat surface A, the flux is:

5BUAUcos 6:3

When the surface is in parallel to the field (5/2), there is no flux throughthat surface. The flux is maximal through a surface perpendicular to the field

(5 0).The magnetic field strength of (permanent) magnets is expressed in terms of

magnetic induction B (so in Tesla) rather than of H. For instance the Earths mag-

netic field strength is about 60 T, and the strengths of permanent magnets rangefrom 0.01 to 1 T.

Free charges moving in a magnetic field experience a Lorentz force, driving

them into a direction according to the well-known right-hand rule:

Fl 5 qv3B 6:4

This happens also with the free electrons in a conductor that moves in a mag-

netic field. The movement results in a potential difference, the induction voltage,

across the conductor and satisfies the equation

Vind 52ddt

6:5

which is the induction law of Faraday. In an open wire loop, moving in a magnetic

field, the induction voltage appears between both ends of the wire; the current

through the wire is zero. In a closed loop, the induced voltage causes a current

equal to Vind/R, where R is the resistance of the wire loop. From Eq. (6.5) it follows

that the induced voltage differs from zero only when the flux changes with time.

At constant flux, the induced voltage is zero. The definition of flux in Eq. (6.5)

links the units volt (V) and Weber (Wb).

6.1.2 Permeability

The magnetic quantities H and B are related by the equation

B5H50rH 6:6

The quantity is the (magnetic) permeability; 0, the permeability of freespace, equals 4 1027 Vs/Am by definition. The relative permeability r is a mate-rial property (compare r for a dielectric material). For vacuum r5 1, for gasesand many nonferrous materials it is very close to 1. The permeability of ferromag-

netic materials is much higher, but strongly non-linear; at higher values of H the

material shows saturation and hysteresis. Table 6.1 shows some values of the static

permeability for several materials, as well as values for the saturation induction.

127Inductive and Magnetic Sensors

• Mumetal is often used for shielding system parts that are sensitive to magnetic

fields. The non-linearity of r is employed in, for instance, fluxgate sensors(Section 6.2.3).

6.1.3 Eddy Currents

Any conductor in a non-stationary induction field experiences induction voltages.

This holds not only for wires (where it is used for the generation of electric cur-

rents) but also for bulk material (as the iron cores of transformers and electric

machines). The induced currents through such material follow more or less circular

paths; therefore, they are called eddy currents. They produce unwanted heat, so

normally they are minimized by, for instance, increasing the resistance of the bulk

material. In constructions with a lot of iron (e.g. transformers and electric

machines), this is accomplished by laminating the material: instead of massive

material the construction is built up of a pile of thin iron plates (lamellae) packed

firmly together. Eddy current can only flow in the plane of the plates, but cannot

cross the boundary between two adjacent plates. A useful application of eddy cur-

rents for sensors is described in Section 6.3.3, the eddy current proximity sensor.

6.1.4 Magnetic Resistance (Reluctance) and Self-Inductance

The analogy between the description of magnetic circuits and electrical circuits is

demonstrated by the equations in Table 6.2.

Equation (6.7) links the intrinsic and extrinsic field variables in the electrical

and magnetic domain, respectively (Chapter 2). The electrical conductivity 5 1/opposes the magnetic permeability . With Eq. (6.8) the field quantities E and Hare converted to the circuit quantities V and I. Equation (6.9) defines density prop-

erties, and Eq. (6.10) defines electrical and magnetic resistances, respectively. The

latter is also called reluctance. Finally, Eq. (6.11) expresses the electric and

magnetic resistances in terms of material properties and shape parameters: l is the

length of a device with constant cross section and A its cross-section area.

In an electric circuit consisting of a series of elements, the current through each

of these elements is the same. Analogously the flux through a series of magnetic

Table 6.1 Permeability of Various Construction Materials [1,2]

Material r (max) Bsat (T)

Vacuum 1

Pure iron 5,000 2.2

Transformer steel 15,000 2.0

Mumetal (Fe17Ni56Cu5Cr2) 100,000 0.9

Supermalloy (Fe16Ni79Mo5) 1,000000 0.8

128 Sensors for Mechatronics

• elements is the same. So the resistances (reluctances) of these elements can simply

be summed to find the total reluctance of the series circuit.

The self-inductance of a magnetic circuit with coupled flux is found as follows.

The induced voltage equals Vind 5 nUd=dt (when there are n turns). Substitutionof using Eq. (6.9) yields Vind 5 n2=RmUdI=dt and since V5 L(dI/dt) the self-inductance is:

L5n2

Rm5 n2U

Al

6:12

So the coefficient of self-inductance (unit Henry, H5Wb/A) is proportional tothe square of the number of turns and inversely proportional to the reluctance.

Several sensors, based on a change in self-inductance and reluctance, will be fur-

ther discussed in this chapter.

6.1.5 Magnetostriction

All ferromagnetic materials exhibit the magnetostrictive effect. Basically it is the

change in outer dimensions of the material when subjected to an external magnetic

field. In the absence of an external field the magnetic domains (elementary mag-

netic dipoles) are randomly oriented. When a magnetic field is applied, these

domains tend to line up with the field, up to the point of saturation. The e

Recommended ##### Inductive Cylinder Position Sensors - Steven Engineering .Inductive Cylinder Position Sensors Drawing
Documents ##### INDUCTIVE SENSORS M12 x 1 - Z-TRAUQz-trauq.com/sensors/SI12.pdf · inductive sensors m12 x 1 cylindrical
Documents ##### 871TM Long Range Inductive Proximity Sensors product profile 871TM Long Range Inductive Proximity Sensors
Documents ##### INDUCTIVE PROXIMITY SENSORS - Proximon Controls Pvt.· INDUCTIVE PROXIMITY SENSORS Inductive proximity
Documents ##### Inductive Position Sensors BIL/BIP - murri.fi tunnistus... · Inductive position sensors are typically
Documents ##### sensors worldwide Inductive Proximity Sensors Photoelectric Sensors Capacitive Sensors Sensors for Cylinders
Documents ##### Inductive sensors IME - Industrial Manufacturing Automation .Inductive sensors IME Dependable for
Documents Documents